[ RadSafe ] Fwd: Basis for Sealed Source Leak Test Detection Level
achris1999 at gmail.com
Fri Jun 17 19:01:51 CDT 2016
It's not an LLD per se, it's an action level (threshold, if you prefer).
The NRC's rule is that, when you find 0.005 uCi, or more, leaking from a
sealed source, you take it out of service (like, then), repair or dispose
of it, and inform them (5-day rule??). So, if you cannot detect at least
0.005 uCi, you have a problem, because you could have a leaker, by
regulatory definition, and not know it. Of course, as some of you have
said, it's easy to get LLD's much lower than that, even with, say, a
pancake G-M tube.
Now, what the rationale for the original value was, I don't know. Mayhap
we should page Paul Frame, or a similar historical guru.
More information about the RadSafe