[ RadSafe ] Incidental Radiation Generating Device Leakage Radiation Limits
Dixon, John E. (CDC/ONDIEH/NCEH)
gyf7 at cdc.gov
Wed Sep 5 12:17:20 CDT 2012
Wow! That's the first time in a long time I have seen a reference to the OLD DOE (Admiral Watkins) Radcon Manual! I am not certain if this topic is addressed in newer DOE regulations. The leakage levels in question may be addressed by certain TN State regulations (series 1200***), but these would be noted as radiation boundary levels, not allowable leakage. The ANSI standard N43.3 might be best used here (it is one of many developed for homeland security applications).
For these kinds of radiation generation devices, some NCRP's might be of use. In particular, NCRP 20 (for cargo scanning systems) might be applicable. I would also try NCRP-147 or 144. Look for the same key words: exempt, shielded, or as installed facility.
I would also research the purview of the Food and Drug Administration. If you go their home page and search for ionizing radiation devices manufactures standards, you might find this leakage limit as applicable prior to the device being purchased for such uses. Non-ionizing radiation generating devices, such as commercial microwave ovens, are included under the FDA's responsibilities.
Hope this helps.
John E. Dixon, CHP
From: radsafe-bounces at health.phys.iit.edu [mailto:radsafe-bounces at health.phys.iit.edu] On Behalf Of roseb at gdls.com
Sent: Tuesday, September 04, 2012 10:03 AM
To: The International Radiation Protection (Health Physics) Mailing List
Subject: Re: [ RadSafe ] Incidental Radiation Generating Device Leakage Radiation Limits
Reference U.S. Department of Energy Radiological Control Manual, DOE/EH-0256T, Revision 1 (1994), https://www.orau.org/PTP/PTP%20Library/library/DOE/Misc/doe_rcm.pdf .
The DOE appears to have recognized the absence of specific federal radiological health standards for certain devices in their Radcon Manual (1994), and referred management and the RCO to applicable ANSI standards.
As you are aware, in the case of devices such as electron microscopes and electron beam welders, Article 365.3 leaves the establishment of radiological controls for such devices to line management AND the Radiological Controls Organization (RCO) (DOE, p. 3-29, pdf p. 93/220).
Given the absence of a specific federal standard for the devices under consideration, and, given that the location where the devices are operated is likely a federal enclave or exclusive federal jurisdiction with respect to state radiological health regulations (state regs might not apply), your reference to the ANSI N43.3 Exempt Shielded Installation criteria dose equivalent limit of 0.005 mSv (0.5 mrem) in any one hour at any accessible area 5 cm (2 inches) from the outside surface of the enclosure (N43.3-2008, para. 5.1.10, p.9, pdf p. 18/79) appears to be a reasonable and feasible radiological control measure that is consistent with the DOE Radiological Health and Safety Policy (DOE, p. i, pdf p. 3/220).
Boyd H. Rose, CM, CIH, CHMM, EI
Sr. Safety and Environmental Engineering Specialist Corporate Radiation Safety Officer General Dynamics Land Systems
38500 Mound Road
Mail Zone 436-10-75
Sterling Heights , MI 48310-3269
Tel: 586 825 4503
Fax: 586 825 4015
E-mail: roseb at gdls.com
"Redmond, Randy (RXQ)" <redmondrr at y12.doe.gov> Sent by: radsafe-bounces at agni.phys.iit.edu
09/04/2012 07:25 AM
Please respond to
"The International Radiation Protection \(Health Physics\) Mailing List"
<radsafe at agni.phys.iit.edu>
"radsafe at health.phys.iit.edu" <radsafe at agni.phys.iit.edu>
[ RadSafe ] Incidental Radiation Generating Device Leakage Radiation
Does anyone know a citation for federal leakage radiation limits for
incidental radiation generating devices such as electron beam welders and
electron microscopes? I have searched and cannot find any. As a default,
I'm using ANSI N43.3 Exempt Shielded Installation criteria of 0.5 mrem at
You are currently subscribed to the RadSafe mailing list
Before posting a message to RadSafe be sure to have read and understood the RadSafe rules. These can be found at: http://health.phys.iit.edu/radsaferules.html
For information on how to subscribe or unsubscribe and other settings visit: http://health.phys.iit.edu
More information about the RadSafe